2.Number of tubes:Four(4)
3.Application area:60 m2
7.UV illumination: ≥428uW/cm2
8.Can be operated with 2 or 4 lamps by pressing Lamp 1 or Lamp 2 button
9.Lamps have side compartments for safe storage
10.Angle of the lamp arms can be adjusted from 30-180°
11.Timer can be set to desired duration starting from 15mins to 1440mins
1.Pls clean the room before using uv room sterilizer.
2.Carefully place the UV Room Sterilizer in the center of the room or on a spot where UV light can beam more things in the room.
3.Arrange everything found in the room to ensure they will be exposed to UV light. Place things that are touched more often closer to the system to maximize sterilization.
4.Plug in and set the timer to the desired duration.The system has a time delay of 15 seconds upon timer setting,this allows the system operator to vacate the room.
The UV light will be on after 15 seconds.Sterilization start .
(We suggest a minimum of 30 minutes to a 30 sq.m.areas.)
5."Lamp 1 "control 2 lamps,'lamp 2"control another 2 lamps.
6.It will generate smell after sterilization.It's perfectly natural.Pls open the window to let the smell out after sterilization.
7.Wipe the lamps with dry cloth after sterilization.Carefully stow the arms in the arms door.
1. Please protect your eyes and skin when it works.
2. In order to keep the efficiency, please operate according to the operating manual.
3.UV illumination less than 70% or more than 2000 hours working should change the lamp.